CAN decode and protocol analysis: RS232/UART? Can save as multiple data formats, such as setting, waveform, referance waveform, CSV, picture.
5 bits digital voltage meter and 6 bits hardware frequency indicator function. 32 kinds of auto measurements with statistics, real-time statistics of maxmum, minimum, standrad deviation and etc.2 sets of digital voltmeter. Support threshold testing, free measurements within the screen. Abundant SCPI remote command control.
Many external interface: USB Host/Device. DSO2C10 and DSO2C15 have no waveform generator function! 1GSa/s (single channel) 500MSa/s (two channels).
Display high frequency and random burr. Average waveform, times: 4, 8, 16, 32, 64, 128. 2ns/div100s/div 1, 2, 5 step by step. ±0.5 div×minimum time base gear. Sample Rate and Delay Time Accuracy.Delta Time Measurement Accuracy (Full Bandwidth)Sample Rate and Delay Time Accuracy. Single-shot, Normal mode ±(1 sample interval+100ppm×reading+0.6ns). >16 times averages ±(1 sample interval+100ppm×reading+0.4ns). ±50ppm(at any interval greater than 1ms). Rising time in BNC position (typical).
8 bits resolution, each channel samples simultaneously. +, -, ×, ÷, FFT. Window: Rectangle, Hanning, Hamming, Blackman, Bartlett, Flattop. In''normal'' or''average'' acquisition mode, the accuracy of 10V/div to 10mV/div is ±3%.
In''normal'' or''average'' acquisition mode, the accuracy of 5mV/div to 2mV/div is ±4%. Note: Bandwidth reduced to 6MHz when using a 1X probe. Edge, Pulse width, Video, Slope, Overtime, Window, Pattern, Interval, Under Amp, UART, LIN, CAN, SPI, IIC. ±5 divisions from the center of the screen.
±4 divisions from the center of the screen. EXT(Only With AWG Model).0.2 div×volts/div within ±4 divisions from the center of the screen. Rising edge, falling edge, rising or falling edge. CH1, CH2, EXT(Only With AWG Model). Scanning line, line number, odd field, even field, all field. 0: low level; 1: high level; X: ignore.
Start, Stop, data, Parity ERR, COM ERR. 5 bit, 6 bit, 7 bit, 8 bit.Interval field, synchronization field, ID field, synchronization error, identifier, ID and data. Start bit, remote frame ID, data frame ID, frame ID, data frame data, error frame, all errors, ACK Error, overload frame.
10000, 20000, 33300, 500000, 62500, 83300, 100000, 125000, 250000, 500000, 800000, 1000000. 4, 8, 16, 24, 32. Start bit, stop bit, No Ack, address, restart, address and data. Frequency, period, mean, peak-to-peak, RMS, minimum, mixmum, rising time, falling time, + width, - width, base, top, middle, amplitude, overshoot, preshoot, rising edge phase difference, falling edge phase difference, + duty, - duty, period mean, PRMS, FOVshoot, ROVshoot, BWIDTH, FRF, FFR, LRR, LRF, LFR, LFF.
Hardware 6 bits frequency meter. Sine, square, ramp, Exp, noise, DC.0.1 Hz to 10 MHz. 0.1 Hz or 4 bits, take the greater one. ±3.5 V, high impedance.
V or 3 bits, take the greater one. 800 (horizontal)480 (vertical) pixels. 16 million colours (24 bits true colour).
Minimum, 1 s, 5 s, 10 s, 30 s, infinite. 100-120VACRMS(±10%), 45Hzto 440Hz, CAT? 120-240VACRMS(±10%), 45Hz to 66Hz, CAT? T, 3.15A, 250V, 5x20mm.40+71 °C (-40159.8 °F). 3, 000m (10, 000 feet). 0.31 g RMS from 50Hz to 500Hz. 10 minutes on each axis. 2.46g RMS from 5Hz to 500Hz.
318 x 110 x 150mm (length x width x height). We will ensure that you are satisfied.